Service Guide. Publication Number (order as manual set) The security code is set to “HP” when the function generator is. Note: Unless otherwise indicated, this manual applies to all Serial Numbers. The Agilent .. For information on using the Phase-Lock Option for the HP A. Microsoft® Windows® included. Agilent A. 15 MHz Function /. Arbitrary Waveform Generator. Note: Unless otherwise indicated, this manual applies to all .
|Published (Last):||9 January 2006|
|PDF File Size:||9.43 Mb|
|ePub File Size:||3.60 Mb|
|Price:||Free* [*Free Regsitration Required]|
Page Only the last six characters are recognized from the front panel, but all eight characters are required. If you continued change to a function that is not allowed with the selected modulation, the modulation mode is turned off.
Notice that the annunciator flashes, indicating that the displayed Burst parameter is for the burst mode. From the front panel, you can save the waveform using one of the following names: The output amplitude is Output Amplitude automatically adjusted and no error is generated if you change the continued output termination.
The following steps show you how to store and recall a state. Chapter 3 Features and Functions Calibration Overview Calibration Message You can use the calibration message feature to record calibration information about your function generator. In amplitude modulationthe amplitude of the carrier is varied by the amplitude of the modulating waveform. Page 18 Chapter 1 Quick Start To store the instrument state To store the instrument state You can store up to three different instrument states in non-volatile memory.
Page 5 Display Annunciators Adrs Function generator is addressed to listen or talk over a remote interface. Use the arrow keys to edit individual digits. Page Chapter 3 Features and Functions System-Related Operations Display Control To speed up the rate at which the function generator can make configuration changes, or for security reasons, you may want to turn off the front-panel display.
Page 71 Chapter 3 Features and Functions Frequency Modulation FM Peak Frequency Deviation The peak frequency deviation represents the variation in frequency of the modulating waveform from the carrier frequency.
The default is Both internal-external.
HP A USER MANUAL Pdf Download.
Chapter 3 Features and Functions Output Configuration Instrument State Storage You can store up to three different instrument states in non-volatile memory. Chapter 3 Features and Functions Calibration Overview To Unsecure for Calibration You can unsecure the function generator for calibration either from the front panel or over the remote interface. When you enable the burst mode, the previous modulation mode is turned off. See the next page if you need to change the power-line fuse. The five built-in arbitrary waveforms can be output at a maximum of 5 MHz.
Chapter 3 Features and Functions Calibration Overview To Change the Security Code To change the security code, you must first manuql the function generator, and then enter a new code. Page Chapter 3 Features and Functions Calibration Overview Calibration Message You can use the calibration message feature to record calibration information about your function generator.
You can issue triggers for bursts and sweeps using internal triggering, single triggering, or external triggering. Chapter 3 Features and Functions Output Configuration Output Frequency As shown below, the output frequency range depends on the function currently selected. Features and Functions You will find that this chapter makes it easy to look up all the details about a particular feature of the function generator.
Page 8 The nanual display will light up while the function generator performs its power-on self-test. This limited test assures you that the function generator is operational. Page 44 Features and Functions After one sweep from the start frequency to the stop frequency, the function generator waits for the next trigger while outputting the start frequency.
HP 33120A User Manual
The function generator is shipped from the factory with a mAT fuse installed. Its combination of bench-top and system features makes this function generator a versatile solution for hp31320a testing requirements now and in the future.
The default frequency is 1 kHz for all functions. Synthesized signal generator including options, pages. Page 48 Chapter 3 Features and Functions Output Configuration Output Frequency As shown below, the output frequency manua depends on the function currently selected.
Chapter 2 Front-Panel Menu Operation A front-panel menu tutorial A front-panel menu mnual This section is a step-by-step tutorial which shows you how to use the front-panel menu. Got it, continue to print. The default is Sine.
Chapter 1 Quick Start To adjust the carrying handle To adjust the carrying handle To adjust the position, grasp the handle by the sides and pull outward. Enter text from picture: This enables you to recall the entire instrument configuration with just a few key presses from the front panel.
Don’t show me this message again. The function generator calculates the average of the data points manial compares the average to zero volts. Notice that the annunciator flashes, indicating that the displayed parameter is for the mode.
Only one interface manuxl be enabled at a hp3310a. Page 46 Chapter 3 Features and Functions Output Configuration Output Configuration This section contains information to help you configure the function generator for outputting waveforms. Chapter 3 Features and Functions Frequency Sweep Sweep Time The sweep time specifies the number of seconds required to sweep from the start frequency to the stop frequency.
The Front-Panel Menu at a Glance T he menu is organized in a top-down tree structure with three levels.
You can use sine, square, ramp, triangle, or arbitrary waveforms for the carrier waveform. This enables you to recall the entire instrument configuration with a single command from the remote interface or with just a few key presses from the front panel. Used only in the internal triggered burst mode. To print the manual completely, please, download it. Page 86 Chapter 3 Features and Functions Frequency Sweep Start Frequency and Stop Frequency The start frequency and stop frequency set the upper and lower frequency bounds for the sweep.